Ask a QuestionBy:
May 05, 2011
how do i calibrate thermocoubler
Place the probe into an ice bath and hold down both the F and C buttons till you see "CAL" on the screen for three seconds. Then release both buttons and the screen will display "Hold". Hold down the atkins key the whole time and "Hold" will be replaced with "..." with the current temp. Keep holding the atkins button and pust the down button till you are at 32 degrees F. Once you are done hold the atkins button till you see "End" on the screen. You should be in normal mode again.
From: Los Angeles
Dec 17, 2013
Why does my screen says low?Can get it out from the screen always say low only.
Hello, it is possible there is a glitch with your current product or other issue going on. We would recommend contacting the Cooper-Atkins manufacturer directly for assistance in resolving this issue. Thank you!